The next generation of the fast FID, the FID600 launched in 2019. This offers a new digital data platform, with the ability to log data to file at 4 kHz, a web-based user interface with real-time data visualisation, and new interfaces such as Ethernet, CAN and engine position plus sample operation over a large pressure range (covering typical turbo exhaust pressures). The performance of the FID sensor itself is upgraded, in terms of metrics such as linearity and oxygen synergism, giving it performance more similar to a traditional bench analyzer — but of course with the ability to make ultrafast measurements. Improved soot filtration allows for longer unattended operation. For more information, download the brochure here.
Current Trending Applications
GDI engine combustion optimization during cold start and catalyst heating phase
Application Note: Combustion instability detection during catalyst heating phase of GDI cold start548KB
Sampling from the intake system for evaporative emissions canister purge or crankcase vent HCs
Application Note: Intake manifold measurement of “found fuel”289KB
Real time measurement of engine-out HC during DPF regeneration
Application Note: HC emissions from a diesel-engined passenger vehicle77KB
Ethanol fuel cold start calibration
Application Note: Cold start PFI gasoline HC emissions3.6MB
In-cylinder sampling of unburned mixture through sampling spark plug
Cambustion Gas Analyzers Brochure
The Flame Ionisation Detector (FID) is the industry-standard method of measuring engine exhaust hydrocarbons. Conventional FIDs have response times of 1-2 seconds, and are typically used to measure "bag emissions" where the concentration changes very slowly.
A need for fast emissions analyzers
During the 1-2 seconds it takes a conventional FID to respond, an engine will undergo many firing cycles. The conditions inside the combustion chamber can vary significantly between successive firing cycles- particularly when the engine conditions are changing rapidly such as during engine start or speed/load transients. Cambustion fast FIDs have a time response as low as 1 ms T10-90. This allows the analyzers to distinguish between two adjacent firing cycles, and even offer information about the variation in HC concentration during a single exhaust stroke.
Benefits of fast emissions analyzers
Accurate measurement of exhaust emissions brings valuable insights into engine operation, and assists calibration engineers in reducing the engine out emissions. This can assist in emissions compliance while reducing after-treatment costs.
The fast FID (sometimes known as an fFID) is carefully designed and calibrated to be linear to well above misfire HC levels (e.g. 42,000 ppm C3 for a complete stoichiometric misfire in a gasoline engine) so that accurate concentrations during these important events can be recorded.
For all gasoline engines (port- or direct-injection) the most challenging phase of operation (from an emissions viewpoint) is the cold start. (Download a pdf presentation about gasoline cold start calibration)
The cold catalyst is initially unable to convert any of the engine-out emissions, and these therefore reach the tailpipe. Since in a cold gasoline engine the vaporization of the fuel is poor, additional fuel (i.e. more than stoichiometric ratio alone requires) must be injected during crank, to achieve an ignitable mixture in the cylinder. The rapid heat release caused by the first firing cycle leads to a rapid vaporization of this excess liquid fuel, and the second firing cycle may easily be rich, leading to high concentrations of unburnt hydrocarbons in the exhaust.
The fast FID's unique ability to measure cycle-by-cycle HC emissions during and after start allows engineers to optimize engine calibration for cold start; maintaining startability while minimizing these highly significant HC emissions. The new FID600 has a engine position port for VRS or Hall effect sensors, to give crank angle resolved emissions.
On gasoline engines, the use of an evaporative emissions canister to trap the HC vapours from the fuel tank requires the canister to be purged in to the intake system. Normally, this is scheduled to occur a few minutes after engine start, but can lead to significant levels of additional fuel entering the combustion chambers. With some additional accessories, it is possible to use the fast FID to sample from the intake (even downstream of the throttle) and measure the actual [HC] which are contributing to the engine’s fuelling. Similar techniques can be used for fuel emerging from the crankcase blow-by gases via the crankcase ventilation system.
Alternative fuels such as ethanol (or a mix of gasoline and ethanol) have different vaporization characteristics and require additional calibration for cold start.
Once the engine is running, transients in speed and load can lead to hydrocarbon "spikes" (brief periods of high emission), since the airflow into the engine can change more quickly than the fuel. Conventional analyzers with time responses of around a second can not resolve these events, but the FID600 and HFR500 offer valuable data about the exhaust HC concentration, on a cycle-by-cycle and cylinder-by-cylinder basis.
The FID600 is available with support for AK protocol and CAN allowing easy integration with the test bench for improved test reliability and reduced workload.
The HFR500 is available with support for AK protocol.
To read more about the application of the HFR500 to engine start and fuel puddle studies visit our Spark Ignition Engines page.
For more information on these and other applications please see Applications & Sample data.
The FID600 brochure many be downloaded here.
The HFR500 brochure may be downloaded here.
Please contact Cambustion for more information and prices.
|Title||Data type||Download File||Size||Last updated|
|Cold start PFI gasoline HC emissions||Application note||hfr01v02 si cold start.pdf||77.14 KB||18-Aug-2009|
|High rpm fuelling control||Application note||hfr02v01 high rpm.pdf||149.57 KB||19-Aug-2009|
|Suitability of HFR500 for ethanol measurements||Application note||hfr06v01 ethanol.pdf||13.2 KB||19-Aug-2009|
|Intake manifold measurement of “found fuel”||HFR08v01 intake sampling.pdf||547.89 KB||02-Feb-2015|
|Gasoline Direct Injection (GDI) HC emissions||Application note||HFR07v01 GDI HC emissions.pdf||128.38 KB||24-Feb-2015|
|HC emissions from a diesel-engined passenger vehicle||Application note||HFR05v01 Diesel.pdf||288.67 KB||24-Feb-2015|
|Cycle-by-cycle AFR measurement of a cold start using an NDIR500 and HFR500||Application note||NDIR03v02 AFR measurement.pdf||442.5 KB||24-Feb-2015|
|Flame Traverse Sampling||Application note||CLD06v01 Fast gas analysis traversing a methane flame.pdf||607.88 KB||22-Sep-2015|
|Combustion instability detection during catalyst heating phase of GDI cold start||Application note||HFR09v01 GDI cold start catalyst heating strategy combustion instability.pdf||543.55 KB||24-Aug-2016|