Introduction to the DMS500
Since 2002 hundreds of DMS500s have been adopted in over 30 countries worldwide for engine research, development and calibration. A unique combination of measurement capabilities; particle size , particle number (PN) and soot mass (PM) all in real time has made the DMS500 the measurement instrument of choice.
With comprehensive test cell integration and raw sampling capability, the DMS500 is ideally suited to rolling road and engine test cell applications. Hundreds of technical publications including SAE papers have used DMS500s for engine research.
Particle emissions are subject to a variety of metrics, including particle mass and particle number.
Particle mass is normally measured for legislative purposes using filter paper measurements. This technique has required continuing improvement in sensitivity of mass measurement devices, since the change in mass of the filter paper becomes progressively smaller as engine calibrations improve.
Filter paper mass is inherently an offline technique- reqiuring collection of soot over a period of time before the final weight can be taken. Since engines are tested on transient drive cycles, such an integrated measurement gives no information about what points in the drive cycle contribute most to the cycle soot emissions. An online mass measurement technique is needed to support this. Instruments such as the AVL Micro Soot Sensor allow direct mass measurment, but have a T90-10% response time of around 1 second. Since transients on engine test cycles are typically sub-second, and the AFR excursions in both Diesel and gasoline engines are typically associated with high particle mass emissions, an instrument with significantly sub second T10-90% response time is desirable. The DMS500 offers a T10-90% response time of just 200ms.
The DMS500's sampling system allows measurement both pre- and post-aftertreatment. For example, sampling upstream of a GPF or DPF, the DMS500 can calculate the soot mass output from the engine presented to the filter. This mass data can be used to optimise the engine out soot mass, and to predict the GPF/DPF soot load. Sampling downstream of the GPF/DPF, the DMS500 can give real-time tailpipe particle mass, showing the combined effect of engine out emissions and after-treatment performance on tailpipe emissions.
The particle counting technique is specified in the legislation. This technique offers excellent sensitivity (for post DPF or GPF measurements where concentrations may be low) and commercially available instruments include the Horiba MEXA 1000 SPCS and the AVL 489 APC.
However, the T10-90% response time of Euro 6 PN Counters is typically greater than 1 second. Due to the highly transient nature of particle emissions, particularly from GDI engines, this restricts their use in calibration of the engine / strategy for compliance, although the final certification testing must be performed with such a system.
The DMS500 system is a non-certification system which uses a different technique to measure particle concentration. The DMS500 offers a T10-90%response time of 200ms, and is correspondingly able to give insights into transient calibration issues which contribute a significant proportion of the total emissions on cycle, while correlating with legislative measurements.
The DMS500 has been adopted by vehicle manufacturers and engine developers worldwide as a key tool to understand and reduce PN emissions, from Diesel, GDI and other fuels.
As particle filters are adopted for both GDI and Diesel engines, the particle size becomes increasingly important. Although the two metrics for particle emissions (mass and number) do not require particle size measurement, the efficiency of filtration of a GPF or a DPF is heavily size dependent. To understand the filtration performance of a GPF or DPF, measurement of the size distribution of the exhaust is required. Assuming a nominal size distribution will lead to errors- for example engine load and EGR both affect the particle size distribution. Although some instruments measure one parameter (e.g. PM) and offer a surrogate output of PN, changing particle size will change the factor between these two quantities, making such assumptions inherently inaccurate.
- Measurement of the particle size / number spectrum from 5nm to 2.5μm
- Measurement of Particle Number (PN) for Euro 6 development
- Measurement of Particle Mass (PM)
- The fastest time response available: 10 Hz data, 200ms T10-90% response
- Raw sampling, pre- or post DPF / GPF
- Test cell integration via AK protocol
- Capable of cold-cell testing
- Capable of high-altitude testing
- Measurement of particle emissions from other sources- e.g. blowby, braking...
Fastest Time Response
The fast T10-90% time response is especially useful for investigation of fast changing concentrations and particle sizes. This is particularly important for engine development especially with Gasoline Direct Injection, where high particulate number emissions tend to be associated with transient conditions, rather than steady state.
Even during steady state conditions, the DMS500 dramatically increases the efficiency and convenience of experimental measurements, reqiuring less time at steady state conditions to make the measurement.
Control and Test Cell Integration
The DMS500 is controlled by a PC via an Ethernet interface, allowing the PC to be located up to 200 m from the instrument.
The user interface includes datalogging to hard disk, and data can be sent over AK interface to the test bench.
Control of the DMS500 over AK from the test bench is also possible, for automated operation.
Data presentation tools for MS Excel are included to assist in visualising the data from the DMS500.
Raw Sampling System
The DMS500 is available as standard with a heated sampling line (191C operation) and two stages of integrated dilution. This enables direct measurement from the engine exhaust with no additional equipment required.
Each DMS500 undergoes a calibration procedure, involving real aerosols, taking full account of particle losses, and even different particle morphologies thus preventing the reduction in accuracy at larger sizes seen in some other analyzers. Traceable calibration certificates for spherical and soot calibrations are provided.
|Title||Data type||Download File||Size||Last updated|
|Number : Size Spectrum Correlation with Particulate Mass Measurements||Application note||dms01v05.pdf||194.03 KB||24-Feb-2015|
|Measurement of Volatile Aerosols with the dms500||Application note||dms02v01.pdf||45.93 KB||24-Feb-2015|
|DMS500 sampling options||Application note||dms03v04.pdf||511.06 KB||24-Feb-2015|
|DMS500 measurement of drug delivery inhalers||Application note||dms04v02.pdf||85.64 KB||24-Feb-2015|
|Measurement of DPF/CRT performance with DMS500||Application note||dms05v02.pdf||1.2 MB||24-Feb-2015|
|Real-time mode finding and lognormal fitting with DMS series fast particulate spectrometers||Application note||dms06v03.pdf||682.83 KB||24-Feb-2015|
|Flame Sampling with DMS500||Application note||dms07v01.pdf||260.74 KB||24-Feb-2015|
|Gasoline Direct Injection Particulate Measurement with the DMS Series||Application note||dms08v01.pdf||919.53 KB||24-Feb-2015|
|Particle number measurement with the DMS Series||Application note||dms09v01.pdf||351.19 KB||24-Feb-2015|
|Ambient aerosol sampling with DMS500||Application note||dms10v01.pdf||295.83 KB||24-Feb-2015|
|DMS500 Installation Guide for Engine Sampling||Application note||dms11v02.pdf||512.72 KB||02-Feb-2015|
|Real-time aerosol size, number and mass measurements from an electronic cigarette||Application note||dms12v01.pdf||120.11 KB||24-Apr-2014|
|Real-time particle size measurements of user triggered electronic cigarette smoke||Application note||dms14v02.pdf||910.84 KB||16-Aug-2016|
|Real-time particle size measurements of heat-not-burn tobacco device||Application note||dms15v01.pdf||870.75 KB||09-Oct-2014|
|Brake particle measurements||Application note||dms17v01.pdf||529.01 KB||08-Mar-2016|
|Nanoparticle emissions from cooking||Application note||dms19v01 Nanoparticle emissions from cooking.pdf||1.94 MB||10-Oct-2016|